Basic specifications (Accuracy guaranteed for 1 year)
Measurement lines Singlephase 2wires, singlephase 3wires, 3phase 3wires, 3phase 4wires (voltage / current measurement range set for each wiring mode)
Measurement items Voltage, Current, Active power, Apparent power, Reactive power, Power factor, Phase angle, Frequency, Efficiency, Current integration, Active power integration, Integrated time, Voltage waveform peak value, Current waveform peak value, Voltage crest factor, Current crest factor, Time average current, Time average active power, Voltage ripple factor, Current ripple factor
Harmonic parameters Synchronization frequency range: 10 Hz to 640 Hz, Analysis order up to 50th
Harmonic voltage RMS value, Harmonic current RMS value, Harmonic active power, Total harmonic voltage distortion, Total harmonic current distortion, Voltage fundamental waveform, Current fundamental waveform, Active power fundamental waveform, Apparent power fundamental waveform, Reactive power fundamental waveform, Power factor fundamental waveform (displacement power factor), Voltage current phase difference fundamental waveform, Interchannel voltage fundamental wave phase difference, Interchannel current fundamental wave phase difference, Harmonic voltage content % , Harmonic current content % , Harmonic active power content %
(The following parameters can be downloaded as data during PC communication but not displayed: Harmonic voltage phase angle, Harmonic current phase angle, Harmonic voltage current phase difference)
Measurement range Voltage range: AC/DC 15 V to 1000 V, 7 ranges
Current range: AC/DC 200 mA to 50 A, 8 ranges
Power range: 3.0000 W to 150.00 kW (Depends on combination of voltage and current range)
Integration measurement
(Integration time up to 10,000 hours) [Current] No.of displayed digits: 6 digits (from 0.00000 mAh, Polarityindependent integration and Sum value)
[Active power] No.of displayed digits: 6 digits (from 0.00000 mWh, Polarityindependent integration and Sum value)
Input resistance (50/60 Hz) Voltage: 2 MΩ, Current: 1 mΩ or less (direct input)
Basic accuracy (Active power) ±0.1% rdg. ±0.1% f.s. (DC)
±0.1% rdg. ±0.05% f.s. (45 Hz to 66 Hz, at Input < 50% f.s.)
±0.15% rdg. (45 Hz to 66 Hz, at 50% f.s. ≤ Input)
Display refresh rate 5 times per second to 20 second (depend on average times settings)
Frequency characteristics DC, 0.1 Hz to 100 kHz
D/A output (02/03 model only) 16 channels (selectable from following items), Level output DC ±2 V, Waveform output 1 V f.s.
Level output, instantaneous waveform output (voltage, current, active power)
Level output (apparent power, reactive power, power factor, or other)
Highspeed active power level output
Functions [Rectification method] AC+DC, AC+DC Umn, AC, DC, FND, Autorange, Average, VT or CT ratio settings, Synchronized control, MAX/ MIN, or other functions
Interface RS232C / LAN standard, (01/03 model also includes GPIB)
Power supply 100 to 240 V AC, 50/60 Hz, 40 VA max.
Dimensions and mass 305 mm (12.01 in) W × 132 mm (5.20 in) H × 256 mm (10.08 in) D, 5.6 kg (197.5 oz)
Accessories Instruction manual ×1, Measurement guide ×1, Power cord ×1
